Yes, this is the easiest way to operate the cell. Simply plug the test cell into a PAT-Channel of the PAT-Tester-x and connect the gas in- and outlet. The PAT-Channel can be placed inside a temperature chamber at 0 to +40°C. For more extreme temperatures, the test cell can be plugged into the PAT-Stand-1 which is connected by a cable to a PAT-Channel outside the temperature chamber. Note that the PAT-Cell-Gas only fits into the PAT-Tester-i-16 when disconnected from gas supply.
Comments are closed.