Test cell for optical characterization in the reflective mode.
The ECC-Opto-10 test cell is an advanced next generation battery test cell. It is designed for operando characterization of electrodes using light microscopy, Raman spectroscopy or XRD in reflection mode.
A newly developed sealing concept utilizing laser-welded glass-to-metal electrode feedthroughs and foil seals substantially increases cycle stability compared to the previous generation. The much more compact and low profile design allows use under a wide range of microscopes. We further optimized the cell design for easy assembly. Dedicated sample holders for side-by-side and face-to-face arrangements of electrodes vastly improve the handling.
The ECC-Opto-10 is connected to the battery tester via 2 mm cell cable with banana plugs. It can be used with the PAT-Tester-x-8 as well as potentiostats and battery testers from third-party manufacturers.
High cycling stability due to improved sealing conceptDedicated sample holders for side-by-side and face-to-face electrode arrangements available Fast assembly and dismantling and easy cleaning of cell components Electrodes are easily accessible for post-mortem analysis Low cell height of 21.5 mm for trouble-free use under many light microscopes Fits well on standard microscope sample stages (76 × 26 mm (DIN ISO 8037-1))
Dedicated sample holders
Improve your workflow and save time with our sample holders for side-by-side and face-to face arrangements of electrodes. Both variants are included with each PAT-Cell-Opto-10 or ECC-Opto-10!
Sample holder (side-by-side)
Sample holder (face-to-face)
Contact discs with 1, 2, and 4 mm opening are available and included with each test cell.
Assembly procedures for side-by-side electrode setup (07/2022)
In this updated video, Dr. Matthias Hahn shows the required steps for assembling the test cell in the side-by-side electrode setup inside the glove box.
Assembly procedures for face-to-face electrode setup (10/2021)
Learn how to assemble the ECC-Opto-10 optical battery test cell in face-to-face electrode setup.
Sample test results
The picture shows two lithium metal electrodes – 20 µm lithium foil laminated on copper foil – beneath the sapphire window of the test cell. The electrodes are embedded into a glass fiber separator soaked with electrolyte. Applying a current of 20 µA for 5 hours makes the lithium to dissolve from the supporting copper foil (right electrode) and to plate as dendrites on the opposite side (left electrode).
The picture shows two graphite electrodes embedded in a glass fiber separator soaked with electrolyte. The graphite electrode on the left is being lithiated from the lithium metal electrode below the separator. From the color change, one can infer the lithiation gradient within the graphite electrode. The right graphite electrode can be used in a follow-up experiment to send lithium ions back and forth between the two graphite strips.